In combination with the corresponding evaporator system, the Hei-VOLUME Distimatic automatic module enables efficient, unattended operation around the clock.
1 Vacuum valve
The Hei-VOLUME Distimatic module is operated via the control panel on the control box. In addition to selecting the mode (sensor- or time-controlled), the level sensor in the rotary flask can be calibrated to the medium to be processed by pressing a button. At the same time, the illuminated button displays a status indicator for the level sensor. In the event of an emergency situation, the system can be stopped by triggering the emergency stop switch.
The automatic filling and refilling of the rotary flask is based on the system vacuum. Due to the negative pressure in the rotary evaporator, new medium is fed in as soon as the valve on the control box opens. A sensor detects when the educt is used up. You are free to choose the vessel from which you want to refill and simply replace it during the ongoing process.
1 Educt container
The condensate collects in a collector vessel with a valve matrix, which replaces the receiving flask. The valve matrix consists of a venting valve, a condensate valve and a vacuum valve. The contents of the collector are pumped out sensor-controlled via a chemical membrane pump on the control box into a container of your choice. The collector vessel is ventilated separately for this purpose. Once the collector vessel is empty, it is evacuated again before the condensate valve to the rest of the system opens again and the distillate continues to be collected. An overflow sensor on the condensate collection container detects its fill level and stops the system as soon as the container is full.
1 Vacuum valve
In the Hei-VOLUME Distimatic 24/7 version, the residue in the rotary flask can be automatically pumped out. To do this, the entire system is ventilated and the concentrate is pumped into a container of your choice using the residue pump. An overflow sensor detects the filling level of the vessel and stops the system as soon as the collecting container is full.
For your safety, an cut-out sensor, which is attached to the glassware, monitors the process and stops the system in the event of a fault.
All sensors of the Hei-VOLUME Distimatic module can be adjusted to the medium to be processed.
The Hei-VOLUME Distimatic automatic module is available for Heidolph benchtop and large-scale rotary evaporators.
Take a look at the innovative successor Hei-VOLUME Distimatic Pro.
Our application experts will find the right solution for your requirements!
Contact us using the request form.
|Hei-VOLUME Distimatic - The clever way of continuous evaporation||Download|
|Hei-VOLUME Distimatic at a glance||Download|
|AWARD-WINNING EVAPORATION SYSTEMS – Rotary evaporators||Download|
Due to the automated feed of only a small amount of medium in the rotary flask, the conditions in the overall system are constant in terms of …
No, media with solids content cannot be processed with the Hei-VOLUME Distimatic automatic module.
There are two modes available for programming the automatic module Hei-VOLUME Distimatic: …